Two New LaserCam products joining an already successful LaserCam-HR camera.
The latest release of the BeamView-USB 4.4 software includes several new features that extend the analytic capabilities of the LaserCam-HR laser beam diagnostic systems. Unique features of BeamView-USB software that were not included in the Digital or Analog BeamView software platforms.
Flat Top Beam Analysis
Six additional calculations are now available with BeamView-USB for flat top beam analysis. These calculations are based upon the ISO 13694:2000 standards. The following calculations will allow for greater flexibility for the analysis of applications involving flat top beam shapes. These BeamView-USB calculations may assist in the analysis of Excimer and Nd:YAG laser customers looking to analyze beam uniformity in the near field. The six new features are:
• Plateau Uniformity
• Flatness Factor
• Edge Steepness
• Beam Uniformity
• Effective Irradiation Area
• Effective Average Power/Energy Density
Adjustable Trigger Delay
An adjustable Trigger Delay feature has been added to BeamView-USB 4.4 allowing for a delay to be added (µseconds) to the default trigger delay of the LaserCam-HR camera. This will assist in beam capture when the laser is fired after an external trigger signal that is shorter than the default camera trigger delay. Depending on the default trigger delay of the camera (150 µsec for the CMOS camera types and 20 µsec for the InGaAs LaserCam-HR) and the repetition rate of the laser, the trigger delay feature assists in assuring that a valid laser pulse is captured on the camera during the cameras exposure time.
The LaserCam-HR and LaserCam-HR-UV have a default integration time of 10 msec and the LaserCam-HR-InGaAs camera is set to a default value of 1 msec for optimum performance.
Coherent has locations across the globe that are available to provide support for any product, service or inquiry.
5100 Patrick Henry Drive
Santa Clara, CA 95054 USA
Please Read Before Continuing
Risk factors: Except for the historical information contained here, many of the matters discussed in this Web site are forward-looking statements, based on expectations at the time they were made, that involve risks and uncertainties that could cause our results to differ materially from those expressed or implied by such statements. These risks are detailed in the “Factors That May Affect Future Results” section of our latest 10-K or 10-Q filing. Coherent assumes no obligation to update these forward-looking statements.